1 pack ( 2 x 96 wells)
Adalimumab (Humira®) ADA
Catalog #: SB-06-024
Adalimumab (Humira®) – ADA ELISA
Catalogue Number: SB-06-024
For the quantitative determination of Adalimumab in human serum and plasma
Notice: This product specification data sheet is for information purposes only. Please contact email@example.com for up to date information.
Product Specifications Data Sheet
|Colorimetric; absorbance at 450 nm with a reference at 650nm.
|Kit ships at 2 to 8°C. Multiple storage temperatures upon arrival.
|Detection limit of 84 ng/mL.
|Tolerance up to 7.3 µg/mL.
|Drug tolerance can be improved with acid dissociation. For further details please contact firstname.lastname@example.org.
|Intra-assay precision ranges from 3 to 7%. Inter-assay mean precision is 12%. Determined by analysis of samples at 500 ng/mL in 6 replicates, on 6 different occasions.
|Rheumatoid Factor (RF) Interference
|If high levels are RF are expected, a sample pre-treatment step may be incorporated into the assay. For further details please contact email@example.com.
|Interpretation of Results
|Results are reported as positive or negative relative to a pre-determined cutpoint. To determine the cutpoint, run 4-10 negative controls and calculate their mean, then add 2* standard deviations to the calculated mean.
|Statistical Cut Point
|We strongly recommend each lab develop their own statistical cutpoint in the appropriate study population using methodologies as described by G. Shankar, et al. (2008). (Recommendations for the Validation of Immunoassays Used for Detection of Host Antibodies Against Biotechnology Products. J. Pharmaceutical and Biomedical Analysis 48:1267–1281). For further details please contact firstname.lastname@example.org.
|Safety Warnings and Precautions
|All reagents and samples should be handled as if potentially hazardous and infectious. It is recommended that all kit reagents are handled by trained laboratory staff and are used in accordance with GLP.
Notice: Somru® ELISA kits are custom-made to order. Each kit lot is tested to ensure that it meets specifications before shipment. Minor adjustments to the preparation of reagents may occur due to lot-to-lot variation. Please always refer to the product sheet enclosed with each kit before performing each experiment.
FOR RESEARCH USE ONLY